This item is no longer available. Suggested replacements. UVAS sc Probe, 1 mm, with SC Controller · Get Quote. Continuous UV Results 1 – 20 of Search results for sc Amtax sc Ammonium Analyzer, mg/L NH₄-N, V, 5 m Filterprobe sc. This item is no longer available. Suggested replacements. UVAS sc Sensor, 5 mm, with SC Controller · Get Quote. Continuously protect plant treatment.
|Published (Last):||2 March 2004|
|PDF File Size:||11.75 Mb|
|ePub File Size:||20.94 Mb|
|Price:||Free* [*Free Regsitration Required]|
Inadequate bubble removal from Increase the signal averaging time to a longer interval. Remove the quick connect fitting and wires and install the threaded plug on the opening to maintain the environmental rating. Printed in the U. Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions. Pull the lamp leads and connector through the hole in the lamp housing. Do not touch the new bulb with bare hands.
Got it, continue to print. To locate the representative nearest you, send e-mail to intl hach. Disconnect and remove the existing wires between the quick connect and terminal block J5, see Figure 21 on page 4.
Maintenance Fuse Replacement The controller contains two mains fuses.
Page 23 Allow at least mm Section 7 Troubleshooting 7. In addition, the passcode also regulates function selection for relay options. Measurement Xfer Transfer 4. Rinse with deionized water before calibrating. Turbidimeter Installation Information Installation 3. Wire gauge less than 18 AWG is not recommended. Index Index Numerics 90 Degree Detector Push in and turn to secure the connection.
Section 2 General Information 2. The time is available only in hour military format.
Hach-Lange SC User Manual | 58 pages
Tug gently to check the connection. Changing The Sensor Name Operation 4. Enter text from picture: The barrier must remain in place unless a by qualified technical personnel qualified installation technician is installing wiring for power, alarms, or relays.
Allow at least mm The passcode is factory set to sc the five digits must be followed by a space to remove the trailing asterisk. Index Sample Line Installation Page 20 Installation 3. The display will flash on startup, when the hold outputs function has been activated, and when the filter function signal average is changed to a different value.
Relay loads must be resistive. Don’t show me this message again.
HACH SC E INSTRUMENT MANUAL Pdf Download.
Section 5 System Startup 5. To print the manual completely, hacch, download it. Pass the cable through conduit and a conduit hub or a strain relief fitting and an available access hole in the controller enclosure.
Problem resolution and fuse replacement should only be performed by qualified service personnel. Page 50 Maintenance a. The turbidimeter sensor must be mounted within six feet of the controller unless an extension cable is used.
See Replacement Parts and Accessories on page 55 for ordering information.
Hach sc100 1720E Instrument Manual
Calibrate Sensor as required by regulatory agency Per agency-dictated schedule. Troubleshooting Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions. Sensor Setup Menu Operation 4.
Controller Sc010 Features Operation 4. The calibration history log shows the gain sv100, the time and date of the calibration, and the initials of the operator performing verification. Page 71 Index Sample Line Installation Drain body and clean the body and bubble trap. Table of Contents Section 1 Specifications Seat insulation against connector with no bare wire exposed.
Pay attention to all danger and caution statements. The controller contains three relays designed for use with high voltage Danger: Page of 72 Go. The frequency will depend on the nature and concentration of dissolved and suspended solids in the sample. Section 10 Repair Service Authorization must be obtained from Hach Company before sending any items for repair.